Frequently Asked Questions - All FAQs
Please select your question category
A User Guide is available under the menu "Documents", in the "Software" section. You can also follow the link below. It will take you directly to the "Software" section.
In Internet Explorer 10 it can lead to errors in the map display. To fix this error please disable in the menu "Tools->Internet Options" on the "Advanced" tab the following two menu items.
Please note, that the new programm-functions of Ver. 18.104.22.168 require, that the pressure-logger has minimum the software-version 2.0 and the reading-unit minimum the software-version 2.0. On pressure loggers with older firmware, it comes to error messages that prevent the reading.
Please use the old Drulo Software (version 22.214.171.124) if you hava a pressure Logger with Firmware 1.4 or below. Please note that this version of the software is only working with 32 Bit Operating-Systems.
If you have pressure loggers with firmware-version 2.0 or higher, you have to use the Drulo programm version starting from 2.0.4.xx
Under the menu item „info reading unit“ in the programm menu „help“ you will receive detailed information about the version of the connected readout unit. Please place the pressure logger on top of the reading unit and connect the reading unit to the computer to get the version of the reading unit. Furthermore, ensure that the parameters for the serial-to-USB connection are set to the correct settings.
All F.A.S.T. GmbH measurement devices uses the same USB interface to the PC, thus, the following descriptions can be used for all devices. This also applies to the serial-to-usb interface adapter cable purchased from us. The driver support is for the following operating systems:
Windows 2000, Windows XP, Windows Vista, Windows 7
Please use this link for the installation manual -> FTDI-Driver-installation manual
Please use this link for the driver download -> FTDI-Driver-Download